• Study Resource
• Explore

Survey

* Your assessment is very important for improving the work of artificial intelligence, which forms the content of this project

Transcript
```Principles of Imaging Science II (RAD 120)
X-ray Imaging System Circuitry
X-ray Imaging System
• Operating console
– Set x-ray tube current (quantity) and voltage (quality)
– Controls line compensation, kVp, mA and exposure time
(mAs) via digital meters
– AEC devices
2
X-Ray Circuit
3
1
Operating Console
• RT controls & meters are located within the low voltage
side (primary) of the x-ray circuit
• Reduce shock potential
4
Operating Console Line Compensation
• Incoming Line Voltage (Mains)
– Electricity supply in US is 60 Hz AC, with a nominal
rms of 200 – 240 volts
• Polarity reverses 120 times/sec
• Voltages constantly fluctuate as resistors are activated
in circuit in accordance with Ohm’s Law
– Can vary +5% affecting x-ray production
– Supplied in the form of a three-phase power cycle
5
Single-Phase Power
• A. Voltage drops to zero
with every change in
direction
– Frequency of sine wave
is determined by # of
cycles/sec (cps)
– 60 Hz X 2 directions =
120 changes/sec
• B. Full wave rectified
circuit (DC pulsating)
– Produces no x-ray
photons 120X/sec
7
2
Single-Phase Power 1o
• RMS V of a single-phase
sinusoidal wave is
approximately 70.7% of
peak voltage
• Calculate the rms voltage of
a single-phase sine wave
with 90 kVp peak?
• 63.6
• 65 kVp peak?
• 46
• Inefficient, solve by using
three-phase power
8
Three-Phase Power 3o
• Supplied by power co
• As each wave peak begins
to drop toward 0, voltage
is boosted back to peak by
next phase
• A. Sum of phasing never
drops to 0
• Produces 3 pulses/half
cycle
– 6 pulses/Hz &
pulses/sec
360
9
X-Ray Circuit
10
3
Autotransformer
– Single winding around an iron core
– Operates on self-induction principle
– Good for controlling voltage on low voltage side of
x-ray circuit
– Supplies precise voltage to the high voltage
(secondary) and filament circuits
– @220 volts delivered to primary side from
incoming line voltage
• Voltage compensation automatic on 1o side
– Follows the transformer law
11
Autotransformer
• A, A1: primary
connections that conduct
input power to
Autotransformer
• C: Increases voltage due
to proximity to end and
number of turns encased
by the connections
• E: Decreases voltage
12
Kilovolt Peak (kVp) Selector
– kVp selector on output side
• kVp major, kVp minor controls
• 220 volts delivered to primary side output voltage of
autotransformer is usually between 100 to 400 volts
• Output voltage is then delivered to input side of step up
transformer for x-ray tube operation
13
4
Milliamperage (mA) Selector
•
•
•
•
•
X-ray tube current is controlled
by the filament circuit
Thermionic emission is based
upon temp of the filament
measured in amperes (A)
Filaments operate at 3 – 6
amps, 6-12 volts
Fixed mA stations as resistors
– Max mA, drops
14
Milliamperage (mA) Selector
• Voltage from mA station is
delivered to filament
transformer (Step-Down)
– Lower voltage, higher
current to filament
• mA meter measures x-ray
tube current
– Placed in the center on
output side of high
voltage transformer
– May be placed on control
console
15
Exposure Timers
•
•
•
•
Determines exposure duration
– Connected on the primary side of the high voltage transformer
Types:
Electronic: Most common, microprocessor controlled.
– Short time 1ms
– Good for multiple sequence imaging
mAs Timer: Electronic timer monitors tube current and is on the
output side of the high voltage transformer.
– Uses the shortest exposure time for mAs selected
– Used in falling load generators
•
•
•
•
•
Designed to work in 3 phase or high frequency generators
Kvp, mA regulated separately
Exposure begins at highest mA, then decreases
Permits better use of acceptable x-ray tube limits; less costly
AEC
16
5
Voltage Rectification
• Required for x-ray tube operation
• Process of changing alternating current (AC) to
pulsating direct current (DC)
• A rectifier functions by allowing current to
flow through it in one direction only
• Electron flow in the x-ray tube must be from
cathode to anode
17
Semiconductors
• Modern method of rectification
• N type & P type semiconductors are used
– N-type have loosely bound electrons that flow
easily between the atom’s conduction bands
• Silicon/Phosphorus, Silicon/Arsenic
– P-type have electron traps (positive holes) that
attract and hold electrons instead of allowing
them to move to another atom
• Silicon/Boron, Silicon/Gallium
18
Diode
• Joining of n-type and p-type semiconductors
• Electrons are attracted toward positive charge
and move through the n-type material to the
junction between the semiconductors.
Additional electrons move in to replace
electrons that migrated
• At the junction, electrons are attracted to the
positive holes keeping a continuous electric
potential in one direction only
19
6
Solid-State Semiconductor Diode
20
Types of Rectification
• Half-wave Rectification
– Single rectifier that
suppresses the negative
half of the alternating
cycle
current
• Energy loss in form of
heat
21
Neg Cycle
Types of Rectification
• Full-Wave Rectification
– Four rectifiers that are
arranged to allow
electron flow from
negative - positive
– Uses all the current flow
from the AC source
• Rise and fall of current
potential
• Rippling of current
produces lower energy
x-rays
Pos Cycle
22
7
Full-Wave Rectification
• Positive Half Cycle
– Diodes A & D permit
electron flow during the
positive half cycle
– Diode C cannot conduct
electrons
• Negative Half Cycle
– Diodes B & C permit
electron flow
– Diodes A & D block
electron flow
23
Three-phase generators
• Incorporates three out-ofphase currents to produce
ripple
• Three coils of wire are
wrapped around the
generator core.
24
High-frequency DC
• X-ray circuitry is designed
to increase the standard
60 cycle frequency to 505,000 cycles/sec
– Nearly constant potential
waveform
– Smaller in design
quality and quantity
– Lower patient dose
– Increase x-ray tube life
25
8
Voltage Ripple
26
X-ray Circuit Diagram
27
9
```
Document related concepts

Switched-mode power supply wikipedia, lookup

Opto-isolator wikipedia, lookup

Current mirror wikipedia, lookup

Power electronics wikipedia, lookup

Voltage regulator wikipedia, lookup

Schmitt trigger wikipedia, lookup

Resistive opto-isolator wikipedia, lookup

Surge protector wikipedia, lookup

Current source wikipedia, lookup

Ohm's law wikipedia, lookup

Spark-gap transmitter wikipedia, lookup

Rectiverter wikipedia, lookup

Multimeter wikipedia, lookup

Power MOSFET wikipedia, lookup

TRIAC wikipedia, lookup

Operational amplifier wikipedia, lookup

CMOS wikipedia, lookup

Valve RF amplifier wikipedia, lookup

Josephson voltage standard wikipedia, lookup

Klystron wikipedia, lookup

Nanofluidic circuitry wikipedia, lookup

Cavity magnetron wikipedia, lookup